If ibtest completes with no errors, you have installed the NI In the Code Editor window, under the Object dropdown list, select Form. Worldwide presence We supply and support our products across the globe, with seven offices in the Americas, Europe and Asia as well as additional representation. This page location is: Get the resources you need.
|Date Added:||25 August 2010|
|File Size:||43.69 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
If the files in steps 2 and 3 are not installed on your computer, they may be copied from the National Instruments setup disks or they may be downloaded from www.
Disconnect any GPIB cables. From success stories and product videosto product literature and newsletters, find the information you need about our switching and simulation solutions.
Cables – Connector to Unterminated 12 Additional Products: Add three Label controls to the form. Add controls to form: Verify that the following files have been installed to the Windows System folder:. Please visit Request Support for product support inquiries.
Legacy: GPIB Hardware and Software Diagnostic Tests under DOS
Resource center From success stories and product videosgpiv product literature and newsletters, find the information you need about our switching and simulation solutions. If ibtest completes with no errors, you have installed the NI Call National Instruments if you feel this is the case.
If the diagnostic continues to fail, then there are only three possible reasons: Cables – Connector to Unterminated 27 Additional Products: If you are writing a new GPIB application we do not recommend using these device templates. Sometimes users accidentally set the switches exactly opposite of what they should be. For details on doing this, return to the previous step: Run the Software Diagnostic Program To verify and test the hardware and software installation, run the ibtest software diagnostic program that came with your NI It is very important that the DIP switches and jumper settings on your board match the settings selected in the GPIB configuration utility.
Type Exit and press Enter to quit.
Legacy: Non-Plug and Play GPIB Hardware for DOS and Windows /95/98/NT – National Instruments
Instead you should make device parameter changes dynamically in your application code. Press any key to continue on to the GPIB device map. COM resides and how many interfaces it supports. Add the code to this subroutine as shown in Table 2.
You must change the DIP switch and jumper settings on the board to match your software configuration. Cables – Connector to Connector 6 Additional Products: Xos Pentium 90 or higher is recommended, running Windows 95 or better.
Verify that the following files have been installed to the Windows System folder: Search our knowledgebase for information on software drivers, product comparisons and technical support for our entire product range.
If you have a jumpered non-plug and play board, NI-PnP will not glib detect that your board is in the computer. Connectors 4 Tools 2 Other Accessories 2 Pin 1.
If ibdiag completes with no errors, your hardware is functioning properly. Connector Blocks 3 Additional Products: If you are using a Gplb and Play board, after selecting an address, change the board’s 5 DIP switches to match the diagram in ibconf. You can now doe proper installation by continuing to the next step: Cables – Connector to Unterminated 9 Additional Products: If ibdiag returns an error message, go back to the previous step and reconfigure the board to not use a DMA channel or an IRQ.
Because we offer such a large range of products, this tool will help you narrow down our offering to get you the correct switch and simulation solution you need.